What are the Steps Involved in Calibrating Pressure Gauge?
March 15, 2018
What is a Dead Weight Tester?
First things first. To understand the calibration process, you first need to know what a dead weight tester actually is. In simple words, it is basically an apparatus, which uses calibrated weights. These weights help check whether the readings shown by the pressure gauges are accurate or not. Fluid, connection port, and a weight and piston are the main components of these testers. The reference pressure of these testers is calculated by the formula –
p = F / A
- p: Reference Pressure (Pa)
- F: Force applied to the Piston (N)
- A: Area PCU (m2)
How is Pressure Gauge Calibrated Using Dead Weight Tester?
To calibrate a pressure gauge using dead weight tester assembly requires one to follow certain steps for a precise calibration. The steps involved in the calibration process are as follows:
Step 1: Making the Arrangements
This is the very first step of the entire process. Before you start the calibration of pressure gauges, make sure that the device is placed at a uniform level. The workbench should be stable for proper operation. Once you ensure this, attach the pressure gauge to the test port.
Step 2: Fixing the Zero Error
You do not want zero error to peek in while calibrating the gauge. Thus, it is advisable to confirm zero reading on the pressure gauge. Adjust the gauge, so that it is reading zero. If this is not done, then the gauge might get calibrated with some zero error still present.
Step 3: Placement of the Weight
This is the step, from which the actual calibration begins. Select a reference weight, and keep it on the vertical piston of the dead weight tester assembly.
Step 4: Moving the Piston
Other than the vertical piston, there is one more piston – a horizontal one. Turn the handle attached to this piston. Do this, until a point is reached, where the piston and the weight are freely supported by oil. Now, turn the vertical piston, so that it starts floating freely.
Step 5: Recording the Readings
Let the system stabilize for some time. Now, once the system gets properly stabilized, it is the right time to record the reading of the gauge and weight.
Step 6: The Final Step
This is the final step of calibration. Follow the steps 3 through step 5, and go on applying pressure. Do not stop until maximum pressure is reached. Now, go on reducing the weights one by one till the point the gauge pressure reads zero. Check whether there is an error in the gauge pressure reading. If the error is within the accuracy limits, then there is no need to worry. Your gauge is calibrated successfully.
A wide range of pressure measuring devices can be calibrated using these dead weight testers. You can easily calibrate industrial pressure gauges, and pressure transmitters with these testers. To know more about pressure gauges, and their calibration, you can get in touch with the experts. The Transmitter Shop is one such experienced player, which can help you calibrate your pressure gauges with great accuracy.
Pressure Monitoring in Pump Systems: A Comprehensive Guide
Common Challenges in Air Flow Measurement and How to Overcome Them
Monitoring and Controlling Energy Production in Power Plants
Understanding the Impact of Pressure Fluctuations on Drying Performance
Understanding Pressure Ranges and Units for Fluid System Monitoring
The Benefits and Challenges of HVAC System Balancing
An Ultimate Selection Guide for Flow Transmitters
Procedure to Calculate Accuracy of Pressure Transmitter Discussed
Pressure Transmitters vs. Pressure Transducers: Learn the Differential Characteristics
Multivariable Transmitter: What Is It and How Does It Work?
How Do You Test for 4 to 20mA Signal in a Pressure Transmitter?
Temperature Transmitter: How to Select The Efficient One for Your Application?
Flow Meter vs Flow Transmitter: Know the Difference
Absolute and Gauge Pressure Transmitters - Overview and Working Principle
HART Communication Protocol: Overview, Working Principle, Benefits in Industrial Automation
What is Absolute Pressure Transmitter & how does it work?
How Do You Calibrate A Flow Transmitter?
Remote Seals: Significance, Working Principle & Applications
How to Select Pressure Transmitter for Your Application?
How to Choose Diaphragm Seals for Your Application?
Difference in Conventional Transmitters and Smart Transmitters
What Are Diaphragm Seals and Their Types?
Rosemount 2088 Vs Rosemount 3051 – A Few Points of Differences Discussed
Rosemount 3051S vs 3051C Transmitter – What is Your Choice?
Impact of Shock and Vibration on Pressure Transducer
Safety Tips for Differential Pressure Transmitter Operation
Factors to Consider When Choosing a Pressure Transmitter Manifold
Tips to Improve the Performance of Pressure Sensors
Important Calibration Tips for Pressure Sensors
5 Most Popular Pressure Transmitter Technologies
Factors of Consideration When Choosing Pressure Transmitters
Tips to Augment the Performance and Service Life of Pressure Transmitter
Factors To Be Considered While Differentiating $40 and $400 Pressure Transmitters
An Unconventional Guide to Selecting the Right Pressure Sensor
3 Major Pressure Transmitter Technologies That Made the Device Popular
The Features and Benefits of Rosemount 1199 Direct Mount Transmitters
All Important Questions on Reconditioned Transmitters Answered
Is Remanufactured Transmitter a Better Option than a New One?
Differential Pressure Transmitters: How Do They Help in Flow Measurements?
3 Whats that Explain How Often You Should Calibrate Pressure Transducer
Guidelines for Troubleshooting Pressure Transducers
Learn How to Calibrate a Pressure Transmitter – II
Learn How to Calibrate a Pressure Transmitter
Know Three Interesting Uses of Pressure Transmitters